Main control board,display panel and power board shall confomi to related drawings.
Labels and printed wordings shall be complete and clear.
Alarm lamp (red): flash to indicate alarming stat
Upper temperature display lamp (orange): lighted when display indicates Upper sensor temperature.
Lower temperature display lamp (orange): lighted when display indicates lower sensor temperat
Upper and lower temperature display lamp both lighted: displays the average value of upper and lower temperature
Power lamp (red): lighted to indicate main power supply connected; flash to indicate main power supply trouble.
Buzz cancel button: to cancel buzzing when alarming ;
Temp shift button: to shift over the displayed temperatures ;
Alarm test button: to test alarm functi With each press, the buzzer beeps 3 times at 1 Hz and alarm lamp flashes 3 times. If remote hardware alarming is not on, remote alarm relay will close for 3s before opens, and then decide if it ' s necessary to alarm; otherwise there' s trouble with alarm ;
Calib confirm button: to calibrate upper, lower and control sensors;
Sensor select button: to select to calibrate sensors;
Calib cancel button: to cancel calibration;
Sensor select button + Calib cancel button: to set temperature (hold for more than 3s).
Circular chart, 7-days temperature recorders
Use pressure sensitive chart paper
No inking is required
Temperature recorder printer
It is very convenient for operating and getting dat
for more information please visit : http://www.hanamedic.com/index.php?page=products&catid=Medical