Overview and performance characteristics:
Image clarity, high efficiency, the use of
international advanced technology Low radiation, the amount of leakage of radiation is only one percent of the national
touch button, microcomputer control, just a button can quickly and accurately determine the exposure parameters
operation room wash, One minute imaging, immediately for diagnosis, the greatest degree of convenience for doctors to diagnose
pneumatic lift seats, more convenient and more comfortable
can be used for oral digital imaging system
Power supply conditions: AC220V ± 10%, 50Hz, 1kVA Tube voltage: 60kVp
Tube current: 8mA Focus size: 1.5mm
Total filtration: 2.5mmAL Exposure time: 0.2-4 seconds
Leakage radiation: 1 meter outside ≦ 0.002mGy / h, National standard 0.25mGy / h)
Optional: tube current: 0.5mA focus size 0.8mm
1. In order to operate the safety, you must install a good grounding wire.
2. Due to the small X-ray machine heat capacity is limited, it is recommended that the operation should pay attention to intermittent work.
3. The machine only in the exposure process, will produce X-ray, power state will not produce X-ray, power on the state as long as the exposure does not produce radiation.
4. The machine in the correct use, any unexpected circumstances, the operator immediately cut off the power, stop using. The aircraft after passing the inspection, before re-use.
5. The factory is equipped with the following specifications of the additional filter board: 1.0 / 0.5mm each one of the aluminum.
6. Remove the machine and remove the other parts when removing the machine.
8. After each use of the machine, the arm to restore the highest point, head position placed balance.
9. After each use with medical alcohol on the end of the beam and the patient contact with the part of the disinfection, cleaning.
10. Clean the machine surface with a dry cloth every week and check that the connectors are in good contact.
11. Attention to the machine maintenance, check the light, display, knob, switch, remote control work is normal; X-ray source components may be oil; X-ray source components exposure, with or without abnormal sound; mechanical parts of the fixed screw There is a loose phenomenon.
12. The rated capacity of the machine is 1.1kW, please use the power supply capacity is greater than 1.5kW power supply or power grid. If the regional grid voltage is not normal, it is recommended to support more than 1.5KW industrial voltage regulator, to prevent the voltage is not normal damage to the machine, reducing machine life.